Because measuring the sulfides in solution is relatively difficult, the Q46S/81 Monitor takes a different approach to the measurement, employing a unique gas-phase method to continuously monitor sulfide without contact between the sensor and the water sample. In operation, a small amount of sample is pumped into the system and mixed with acid. In acidic solution, the sulfide ion is converted to hydrogen sulfide gas as follows:
HS- + H+ H2S
S= + 2H+ H2S
The mixed sample flows into a special chamber where the hydrogen-sulfide is stripped from the sample. A gas conditioning module prepares the gas sample for measurement prior to contact with a special H2S gas sensor designed for the system. Sensor signals are amplified and displayed on a large-format, backlit LCD display in the Q46S electronics unit.
Sulfide ion is measured selectively by conversion to hydrogen sulfide.
Gas Phase Sensing
Measurement is made without contact between sample and sensor, eliminating the potential for sensor fouling.
Chemistry Module Power Options
Power options include 115 or 230 VAC, 50/60 Hz.
Three Control Relays
Relays are programmable for setpoint, deadband, and time delay.
Sample Line Cleaning
Internal sequencing and relay system for automated sample line cleaning.
Communication options for Profibus-DP, Modbus-RTU, or Ethernet-IP.