Click on any of the parameters to learn more.
ATI’s Q46H/79S uses iodometric chemistry, but with a unique sensing technique for measuring released iodine. The system takes the reacted sample containing iodine and uses an air-stripping system to remove molecular iodine from solution.
Employing an Auto-Chem module to provide chemical treatment of the sample, air stripping of the released iodine, gas sample conditioning and iodine gas measurement, the gas-phase iodine from the water sample is channeled through a conditioning module and then directly to an iodine gas sensor. The result is that the iodine measurement is made without any contact between the water sample and the sensor. Contaminants in the sample that cause fouling and contamination of standard electrodes do not affect the Q46 system, providing greater operational reliability
|Q46H/79S Total Chlorine Air Stripping Unit Manual|
|Q46H/79S Total Chlorine Air Stripping Unit Profibus|
|Q46H/79S Total Chlorine Air Stripping Unit Modbus TCP|
|Q46H/79S Total Chlorine Air Stripping Unit Modbus RTU|
Our team of experts are on hand to deal with any queries you may have about our products or service.Get in touch
Accurately controlling a dose of sterilant to clean pipework in the food and drink industry is vital to the taste, quality…Read
Badger Meter and Coftec are pleased to announce the signature of a new partnership, bringing together leading…Read