Click on any of the parameters to learn more.
Continuous water quality monitoring of permanganate in treated water presents sensor contamination issues. When permanganate reacts with organics or other reducing materials in solution, manganese dioxide (MnO2) is formed and readily plates out on sensing electrode surfaces. The resulting deposits degrade the measurement and can be difficult to clean. The Q46/83 eliminates this problem by employing a measurement method in which the sensor never comes into contact with the sample. In operation, water containing permanganate is mixed with pH buffer and potassium iodide solutions. Permanganate oxidizes the iodide to iodine (I2), and the resulting I2 is stripped out of solution and measured using an I2 gas sensor. This “gas phase” measurement technique eliminates MnO2 sensor fouling, resulting in a system capable of providing long term reliability.
The Q46H/83 Permanganate water quality monitor consists of two main components, an electronic display unit and a chemistry module. The chemistry module contains sample and reagent pumps, sample handling systems, and the gas sensor that provides the final measurement. An inlet overflow assembly attached to the bottom of the chemistry module is where the inlet sample line and drain line are connected. Sample inlet flows of 250-100 ml./min. are recommended to keep system response time to a minimum. Reagent bottle holders are supplied so that reagents can be wall mounted below the chemistry system.
The Q46/83 Permanganate Monitor is available for a wide range of applications including:
|Q46H-83 Auto-Chem Permanganate Monitor|
|Q46H/83 Permanganate Monitor Profibus|
|Q46H/83 Permanganate Monitor Modbus TCP|
|Q46H/83 Permanganate Monitor Modbus RTU|
Our team of experts are on hand to deal with any queries regarding our products or services.Get in touch
United Utilities award ATi with a framework for single-source Chlorine, Turbidity, Suspended Solids, Dissolved Oxygen, Sludge Blanket, UV254 and Multi-parameter Network Quality.Read
ATi’s Derek Leslie, Senior Technical Support Analyst, discusses how efficiencies and innovation can optimise water supplies during these drier months.Read
Service reservoir performance analysis requires accurate and reliable monitoring, detailed understanding, and the expertise of a multi-disciplined team.Read